The C3000, C3000MP, C24000 and C24000MP are precision OEM syringe pump modules designed for automated instrumentation. The C3000 and C24000 pumps accommodate valves with 3 or 4 ports in a variety of orientations and the C3000MP and C24000MP pumps support 6-way distribution valves while providing all of the same functionality as C3000 and C24000. All C-Series pumps can be customized for your liquid handling needs.
The C3000 is a seamless replacement for industry standard pumps using the Cavro® protocol. All normal aspiration and dispensing operations expected from an OEM syringe pump module can be accomplished with this unit. The stepper motor driven syringe will accurately and precisely handle fluids from a few microliters to 12.5 milliliters. The syringe drive arm can be easily back driven, making syringe installation or replacement extremely quick and easy.
The C3000MP offers all of the features of the C3000 but is designed to support a 6-way distribution valve.
The C24000 is a high-resolution version of the C3000 that achieves full syringe resolution to 192,000 micro-steps. The C24000 is ideal for low flow rate and low pulsation requirements.
The C24000MP offers all the features of the C24000 but is designed to accommodate a 6-way distribution valve.
The C-Series syringe pumps utilize a proprietary high-efficiency, direct-drive design that does not use belts that can stretch or wear out. The robust drive mechanism has been life-tested to several million cycles with no maintenance required throughout its life.
The C-Series Syringe pumps are ideal units for experienced IVD instrument manufacturers who are looking to maximize the investment they have already made in existing instrumentation. They utilize standard communication protocols and mounting configurations used by other compact syringe pump designs, thereby eliminating expensive and time consuming re-development.
ALL MOUNTING HOLES ARE M3 X 0.5mm X 6.3 mm DEEP MINIMUM
ALL DIMENSIONS UNLESS OTHERWISE SPECIFIED ARE .XXX + .008 [.02 mm]
*S/N 336945 or earlier may require firmware upgrade.
** 6-way distribution valve is only compatible wtih C3000MP and C24000MP syringe pumps (Cat# 10095 & 10099).
TriContinent operates an ISO 13485 registered manufacturing facility that delivers the highest quality products. C-Series Pumps are general laboratory components not subject to FDA regulatory approval. TriContinent uses the highest quality components with applicable safety markings to meet the needs of system designers. C-Series Pumps do not carry independent CE markings as they are components designed for use within larger systems which are subject to independent testing and certification.
We stand behind the C-Series Pumps with a 1-year warranty. During the warranty period, should a C-Series Pump fail, call us with an explanation of malfunction and TriContinent will issue a Return Material Authorization (RMA) number for repair or replacement at our discretion, at no charge.
TriContinent also guarantees its Valves to be free of defects in materials and workmanship for 1-year. Any valve, which fails due to such defects, will be repaired or replaced at our dscretion at no charge provided the device is returned through our RMA process.
It is the responsibility of the customer to determine the suitability of an application and the material compatibility of our product with your application. When returning a product for repair or replacement, please contact TriContinent Customer Service to receive an RMA number. When calling please be sure to have the following information available:
TriContinent will not accept the return of any product with out prior approval. TriContinent reserves the right to refuse a return of product that has been used with infectious microbiological or radioactive substances or any other materials that may be deemed hazardous.
The RMA number should be referenced on the outside of the shipping container when returning to TriContinent.
The above warranty shall not apply for defective products resulting from:
There is no warranty expressed for the following products:
Note: For equipment damaged in shipping, please contact the shipper directly.